Eur. Phys. J. Appl. Phys.
Volume 27, Number 1-3, July-September 2004Tenth International Conference on Defects: Recognition, Imaging and Physics in Semiconductors (DRIP X)
|Page(s)||503 - 506|
|Section||Electronic properties through contactless characterisation|
|Published online||15 July 2004|
Mapping of minority carrier diffusion length and heavy metal contamination with ultimate surface photovoltage method
Semiconductor Diagnostics Inc. 3650 Spectrum Blvd.
#130, Tampa, FL 33612, USA
Corresponding author: email@example.com
Accepted: 13 February 2004
Published online: 15 July 2004
The Ultimate surface photovoltage method of minority carrier diffusion length measurements reffered to as Ultimate SPV, replaces a sequential “one wavelength at a time” approach with simultaneous illumination with the entire set of wavelengths. In this multiwavelength beam, each monochromatic component is chopped with slightly different frequency. This enables simultaneous monitoring of all component SPV signals corresponding to different wavelengths using multi-frequency signal processing. The amplitude and phase of each component signals are then analyzed and used to calculate the diffusion length and surface lifetime. In-flight Ultimate SPV measurement, whereby the wafer continuously moves under SPV probe, is used for fast whole wafer mapping. In addition to speed advantages, Ultimate SPV offers a fundemental accuracy advantage due to elimination of differences in wafer condition during sequential illumination with individual wavelengths. High-speed measurements make it possible to add additional wafer treatments and perform multi-mapping required for separation of Fe and Cu in the silicon bulk. Wafer mapping in time of 2 minutes realized with Ultimate SPV is critical for monitoring of cobalt in silicon.
PACS: 72.40.+w – Photoconduction and photovoltaic effects / 72.20.Jv – Charge carriers: generation, recombination, lifetime, and trapping
© EDP Sciences, 2004
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.