Eur. Phys. J. Appl. Phys.
Volume 33, Number 1, January 2006
|Page(s)||59 - 67|
|Section||Instrumentation and Metrology|
|Published online||14 December 2005|
A half-analytical formulation for the impedance variation in axisymmetrical modelling of eddy current non destructive testing
Laboratoire de Génie Électrique (LGE), Université A. Mira,
route de Mézaïa Targa-Ouzamour, 06000 Bejaïa, Algeria
2 IREENA, LRTI-IUT-CRTT, boulevard de l'Université, BP 406, 44602 Saint-Nazaire Cedex, France
3 Université F. Abbass, Cité Mabouda, 19000 Sétif, Algeria
Corresponding author: email@example.com
Revised: 20 September 2005
Accepted: 30 September 2005
Published online: 14 December 2005
We present the calculation of the impedance variation using a half-analytical formulation based on coupled electromagnetic variables. Such a formulation concerns an axisymmetrical device constituted with a voltage supplied solenoïdal inductor and a conducting workpiece. In this field of modelling, authors have already developed a method [Maouche and Feliachi, J. Phys. III France 10, 1967 (1997)] that determines the current distribution inside inductor coil loops in the case of weak skin depth and a low number of these coil loops. In the proposed development, the number of loops is relatively large and the skin effect in these loops is negligible. This formulation uses a voltage excitation, which makes the source field depending on induced currents and permits to consider the real geometry of the inductor. The model is applied to study an eddy current non destructive testing (ECNDT) device. The variation of the system impedance is calculated in the case of an axisymmetrical device. The obtained modelling results are validated by comparison to measurements and finite element computations [Rémy, Ph.D. thesis, University of Compiègne, France, 1997; La et al., Rev. Prog. Quant. Non-Destructive Eval. 16A, 295 (1997)]. Once validated, the proposed model is applied to determine geometrical and physical characteristics of an ECNDT device. To assemble this interest, we visualise the evolution of the impedance variation according respectively to the air-gap, to the thickness of the workpiece and its electric conductivity. The model is implemented within a software tool (CECM: Coupling Electromagnetic Circuits Method) developed in Matlab environment.
PACS: 81.70.Ex – Nondestructive testing: electromagnetic testing, eddy-current testing / 85.70.Ay – Magnetic device characterization, design, and modeling
© EDP Sciences, 2006
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.