Eur. Phys. J. AP
Volume 15, Number 2, August 2001
|Page(s)||141 - 147|
|Section||Instrumentation and Metrology|
|Published online||15 August 2001|
A virtual non contact-atomic force microscope (NC-AFM): Simulation and comparison with analytical models
Centre de Physique Moléculaire Optique et Hertzienne (CPMOH), Université de Bordeaux I, 351
cours de la Libération,
33405 Talence Cedex 05, France
Corresponding author: email@example.com
Revised: 9 April 2001
Accepted: 9 April 2001
Published online: 15 August 2001
A virtual NC − AFM machine has been built using the Matlab language. The virtual NC − AFM is identical to a real hybrid machine built with Digital Instruments and Omicron blocks. The role of every subset is described in detail, and special attention is paid to the parameters of the Automatic Gain Control (AGC) which controls the amplitude of the oscillations. The virtual machine is a powerful tool, that allows to solve, without any approximation, non linear coupled differential equations describing the physics of the tip-surface interaction. The machine is then used to study the frequency shift and damping signal in the approach-retract mode. Two types of situations are analyzed: the first one corresponds to the case where no dissipative force is introduced in the tip-surface interaction, the second one deals with dissipative forces. These last ones are here introduced by assuming a mechanical deformation of the surface under the tip. Interesting results are then obtained, in particular the fact that unstability may occur under some particular conditions. Most of the results obtained by simulation are then compared with those of analytical models already published in the literature.
PACS: 68.37.Ps – Atomic force microscopy (AFM)
© EDP Sciences, 2001
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.